If you do not follow the instructions concerning our policy on external links
your submission will be sent to the spam folder.
Reason for Recall:
Becton Dickinson (BD) is recalling Alaris Infusion Sets, due to the potential for faster than expected delivery of medication (over-infusion) or an unintended delivery that occurs while the pump is not in a "running status." The firm has determined that the silicone segment of the affected administration set has non-uniform thickness. Non-uniform wall thickness can lead to non-uniform tubing collapse and can contribute to a failure to fully occlude the tubing. This device defect may cause serious adverse health consequences for patients, including death. This recall has been associated with MDR reports, several of which are associated with serious injuries.